---
product_id: 69890194
title: "Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C"
brand: "sinometer"
price: "1935238₫"
currency: VND
in_stock: false
reviews_count: 7
url: https://www.desertcart.vn/products/69890194-sinometer-10-mhz-single-channel-oscilloscope-cq5010c
store_origin: VN
region: Vietnam
---

# 10 MHz Bandwidth 5mV/DIV Sensitivity Low Power Consumption Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C

**Brand:** sinometer
**Price:** 1935238₫
**Availability:** ❌ Out of Stock

## Summary

> 🔍 Discover Precision Like Never Before!

## Quick Answers

- **What is this?** Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C by sinometer
- **How much does it cost?** 1935238₫ with free shipping
- **Is it available?** Currently out of stock
- **Where can I buy it?** [www.desertcart.vn](https://www.desertcart.vn/products/69890194-sinometer-10-mhz-single-channel-oscilloscope-cq5010c)

## Best For

- sinometer enthusiasts

## Why This Product

- Trusted sinometer brand quality
- Free international shipping included
- Worldwide delivery with tracking
- 15-day hassle-free returns

## Key Features

- • **User-Friendly Design:** Includes a probe and manual for seamless setup and operation.
- • **Comprehensive Warranty:** 90-day money-back guarantee and 1-year limited warranty for peace of mind.
- • **Lightweight & Portable:** Easily transportable for on-the-go professionals.
- • **Versatile Operation Modes:** TV sync and X-Y operation for diverse applications.
- • **High Sensitivity Performance:** Capture even the smallest signals with 5mV/DIV precision.

## Overview

The Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C, is a lightweight and efficient tool designed for professionals seeking high performance and sensitivity in signal measurement. With versatile operation modes and a robust warranty, it’s the perfect companion for both fieldwork and lab environments.

## Description

Vertical Deflection Operating Mode CH1 (Y1),CH2 (Y2), CHOP, ALT, ADD, X-Y Sensitivity and Accuracy (Both Y1 & Y2) 5mV-10V/div, 5X MAG, in 12 calibrated steps of 1:2:5 sequence Band-width DC: 0 ~ 20Mhz / AC: 10Hz ~ 20Mhz Rise Time 17.5ns Input Coupling AC, GND, DC Input Impedance 1M ohm (<30pF) Maximum Input Voltage 400V (DC+AC peak) CH1 Signal Output 25mV/div, 50 ohm, 20Hz-10MHz(-3dB) Horizontal Deflection Sweep Mode AUTO, TRIG, LOCK, SINGLE Time Base 0.1 uS-0.2S/div, 20 steps of 1:2:5 sequence Sweep Magnification 5 X (Maximum sweep speed is 20 ns per division) Accuracy 3%, 5% at 10X times Trigger Source CH1, CH2, ALT, Ext, Line Trigger Coupling AC/DC, TV/NORM Trigger Slope Rising or falling edges External Trigger Input Impedance 1M ohm Minimum trigger level 5Hz ~ 20MHz: INT - 1 div; EXT - 0.2Vp-p TV Signal: INT - 2 div; EXT - 0.3Vp-p Trigger Lock: INT - 2 div X-Y Operation Input X axis: CH1 Y axis: CH2 Frequency Response DC: 0 ~ 1 MHz (-3dB); AC: 10Hz ~ 1 MHz (-3dB) Phase Difference <3 degree General Power AC110V, 60 Hz Power Consumption 35W Dimension 316x132x410(mm) Dimension 316x132x410(mm) Weight 16 lbs

## Features

- Lightweight with low power consumption
- TV synchronizing, X-Y operating
- Supplied with 1 (X1, X10) probe and manual
- High performance & sensitive 5mV/DIV
- 90 days money back guarantee and 1 year limited warranty

## Images

![Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C - Image 1](https://m.media-amazon.com/images/I/71P29vQYfaL.jpg)
![Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C - Image 2](https://m.media-amazon.com/images/I/518HRF14HML.jpg)
![Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C - Image 3](https://m.media-amazon.com/images/I/71lAiG9x2CL.jpg)
![Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C - Image 4](https://m.media-amazon.com/images/I/51vRmoJBF7L.jpg)
![Sinometer 10 MHz Single Channel Oscilloscope, CQ5010C - Image 5](https://m.media-amazon.com/images/I/71tya0GZNbL.jpg)

## Customer Reviews

### ⭐⭐⭐ 3.0 out of 5 stars







  
  
    the YB4328 I got had some problems
  

*by P***R on Reviewed in the United States on July 12, 2017*

I've had this scope for 2 years now before reviewing it.  I had some problems with it.  First, after a couple of weeks it started to have an electrical smell I associate with electronic equipment of the 1960's burning in.  Also one probe was jest about impossible to compensate. The scope came without a compensation tool too and I didn't have one.  Also channel  2 with signal would move up and down a half a division for a while once in  while.  It would do that on "ground " signal too.  The real problem game later. After about 6 months both channels would start to degrade  significantly.  Digital signals would have very rounded edges from about half way up or down on the edges. Slightly moving the V/div knob back or forth I could pin the signal to behave normally with nice square edges.  It got worse.  Eventually I sent it to the seller for repair under warranty.  My shipping cost 1/3 the price of the scope.  It lasted a few months, then started up again little.  The seller had told me cleaning the switch contacts is all they had done  (which I had suggested!)  I used it about another year living with it and it did not get really bad again.  Recently, the voltage scale switches got cleaned again without costing me anything (neat trick but I can't say that is readily available).  The other probe had frozen too.  Having been not practicing electronics for over 20 years I can say I have had many hours enjoying the use of the scope.  I wanted a low cost analog scope and this is what I found.  However, the problems I had seem significant to me and I feel I should mention them.  The problems all seem unrelated to each other.  It may be this unit is just particularly bad but I'm unlikely to buy Sinometer again.  The seller I felt was minimally satisfactory.  The didn't have a replacement unit the could send me.  Having said that the front design is nice.  The other switches all work.  I could pick out other performance flaws but for a low cost analog scope what can one really expect?.....there does seem to be a low frequency problem on the front end,  signals that should be square sometimes slant downward to the right.  some the edges on signal seem to overshoot in such a way I believe it's the scope.  Lastly I'll mention that even though a 10v source will measure as though in calibration, 10v,  a signal may measure 10.4v, enough to notice.  I'm not sure why a beam would spread.  I wish 3 stars as a recommendation was considered neither good nor bad, rather than it's okay.  I feel 3 stars is generous with the warnings I have stated about problems.  I have not seen any other review of the YB4328.

### ⭐⭐⭐⭐⭐ 5.0 out of 5 stars







  
  
    Excellent accuracy for the price
  

*by M***K on Reviewed in the United States on August 17, 2014*

The manual, though lacking examples, was pretty clear to understand - even for an oscilloscope newbie.  I think the fact that this is an analog product has allowed them to offer an oscilloscope with very fine precision at an affordable price.  I had anticipated that the product would be quite heavy, but the CRT is very small. It is actually fairly light for its size. However, it does appear to be most likely fairly fragile. This is fine for me, as I do not plan to move it. But if I were to take this offsite I would be packing it appropriately. In any case, once I calibrated the unit and understood how to use, I was able to see the source of the issue I bought it to solve. As an FYI I am working with low voltage DC electronics. So certainly for my purposes so far this item has fit the bill, and so far at least I would recommend. The one thing to note though is that while the oscilloscope has both X and Y axis inputs, there is only 1 probe included. Perhaps there is a reason I do not understand yet but it is something to be aware of.

### ⭐⭐⭐ 3.0 out of 5 stars







  
  
    It actually works
  

*by J***N on Reviewed in the United States on December 20, 2019*

Buttons are a bit soft, but this scope does the job.  I was a bit dismayed to find in XY mode the X channel was AC coupled only, which means traces didn't stay where I wanted them, so I modified the input (yes, opened it up and added a 2k resistor across the input coupling capacitor).  Made another modification to add in a Z-axis input by modifying blanking amplifier and mounting a BNC to back panel.  That works fine now too.  So I got what I was looking for at a decent price.  Better than taking a chance on eBay...

---

## Why Shop on Desertcart?

- 🛒 **Trusted by 1.3+ Million Shoppers** — Serving international shoppers since 2016
- 🌍 **Shop Globally** — Access 737+ million products across 21 categories
- 💰 **No Hidden Fees** — All customs, duties, and taxes included in the price
- 🔄 **15-Day Free Returns** — Hassle-free returns (30 days for PRO members)
- 🔒 **Secure Payments** — Trusted payment options with buyer protection
- ⭐ **TrustPilot Rated 4.5/5** — Based on 8,000+ happy customer reviews

**Shop now:** [https://www.desertcart.vn/products/69890194-sinometer-10-mhz-single-channel-oscilloscope-cq5010c](https://www.desertcart.vn/products/69890194-sinometer-10-mhz-single-channel-oscilloscope-cq5010c)

---

*Product available on Desertcart Vietnam*
*Store origin: VN*
*Last updated: 2026-03-22*